SUN-590 Syringe & Infusion Pump with Drug Library
Syringe pump & infusion pump SUN-590 basic Parameter
Battery Continuous Use Time
About 6 hours
(fully charged new battery running with rate 5ml/h)
AC 100V~AC 242V,50/60HZ(±1 HZ)
DC 12V 1800mAh, Ni-MH, rechargeable
Infusion Flow Selection
0.1ml/h~1200ml/h (1-150d/min), original value is 0.1ml/h (1d/min)
Syringe Flow Selection
0.1ml/h~1600ml/h, original value is 0.1ml/h
5/6ml, 10/12ml, 20/25ml, 30/35ml,50/60ml
Syringe 30ml:900 ml/h
Syringe 20ml:600 ml/h
Syringe 10ml:400 ml/h
Syringe 5ml: 200ml/h
0.1ml~9999.9ml, original value is 0.1ml
volume mode ±5%, drop mode ±2d
Syring Flow Rate Precision
High:110kPa±20kPa (825mmHg±150 mmHg)
Low:60kPa±20kPa (450mmHg±150 mmHg)
The biggest output pressure
Relative Humidity:≤90%RH (Under without condensation status)
Electric Shock Protection
Class I, Type CF
occlusion, nearly finish, finish, standby (no operation),
door open, low battery, battery exhausted etc.
power indicator, transfusion amount display, flow rate
display, infused volume display etc.
After finishing predetermined infusion volume, to prevent the formation of thrombus, continue infusion with preset minimum rate 0.1-5ml/h(setted).
Power Supply Switching Function
When the AC power supply is off, it will automatically switch to
internal battery power supply and prompt at the same time. When
AC is switched on, it is automatically converted back to AC supply.
Repeat Alarm Function
After the operator presses the “” key, if the reason causing the
alarm has be untreated for about 2 minutes, the alarm will sound
again to alert the operator to solve the problem.
Data Memory Function
Save the last setting of sound volume level, brand and occlusion
pressure level, to prepare for the next use. The longest memory time
is 5 years.
For convenience, when installing or removing IV line at night or in
darkness, light will automatically turn on after the door is open. The
light is an LED light source (white).
Battery Protection Alarm
In order to protect the battery and ensure infusion safety, there are
different levels of acousto-optic prompts when the electric quantity
of the battery is changing from close to depletion to almost depleted.
When user tries to remove or take away the syringe, give move alarm.
High Flow Safety Confirm
In order to ensure infusion safety, when the flow rate is greater than
300ml/h (or equivalent), the device will prompt user to confirm
again, and will not carry out infusion before confirmation.
A continuously large number of bubbles going into the human body
can form the air embolus, and it is life-threatening. So when bubble
length is about 3mm (10.6ul) or accumulated bubble length is about
3mm (10.6ul) is detected, will trigger acoustic-optic alarm.
Drip Sensor Slope
In d/min mode, when there are problems such as tube slope, liquid
overflow and misplaced/moved sensor, there will be an alarm.